CAN VOLTAGE AND TEMPERATURE BE RESET BACK TO ZERO BEFORE MEASUREMENT IS TAKEN
Q: Can Voltages and Temperature be reset back to zero before measurement is taken?
Yes you can, you can use the “Perform Auto Zero ADJ” function but this function will only work for Voltage.
STEP 1: ON the GLXXX unit, click on the “MENU” button
STEP 2: When the MENU screen pops up, stay on the first tab, which is AMP.
STEP 3: Scroll down to the channel you want to perform this action to.
- Each channel is broken into categories, such as INPUT, SENSOR, FILTER, EU, Annot and MISC.
STEP 4: Click on the MISC arrow on the channel you want and you will get a new menu.
- Here you can chose what point you want to set zero as and then choose the “Perform AUTO ZERO ADJ” to set that point. Then click ok at the bottom of the menu to finalize your settings.
- After hit the QUIT option to get back to main screen.